Challenges in simulating power supply noise
When you test PSRR, one challenge is modulating a quantitative noise signal onto a DC power supply level. This task is difficult because the power supply inputs typically present extremely low input impedance to any AC signal because of the high input parallel capacitance. While high-performance power supplies do have built-in arbitrary waveform capability, the output bandwidth of power supply waveforms is typically limited to less than 30 KHz. This limitation only allows you to simulate low frequency noise sources, such as power line ripple. It falls well short of the more critical noise source frequency ranges, such as power supply switching noise which ranges from 50 KHz to greater than 20 MHz.
To put the first challenge in perspective, let’s consider an example transceiver chip for a smart phone. The power supply input most likely has a parallel 1-uF bypass ceramic capacitor as well as a large 10-uF electrolytic capacitor to serve as an energy reservoir for the sharp transmit transients. That means a 1-MHz sinewave modulated onto the power supply level of the RF power amplifier would, in theory, only see a load of less than 20 mOhms. In practice, the equivalent series resistance (ESR), equivalent series inductance (ESL), and other parasitic impedances boosts the load impedance up to more than 50 mOhms, but that still presents low input impedance to any AC signal content.
Let's look at two solutions for simulating power supply noise that use (for the most part) general-purpose test equipment and simple components that you can find around the office or lab.
The op amp/MOSFET solution (OMS)
The OMS solution can be seen in the below figure. The solution requires two power supplies, a function/arbitrary waveform generator (FG/AWG), an op amp, and a high-power N-type MOSFET. In Figure 1, C1, C2, and the resistance value labeled “Load” make up a circuit representation of the device under test’s power supply input.
The FG/AWG’s output creates the amplitude level and noise signal at the DUT's input. The power supply output voltage should be set comfortably higher than the desired amplitude plus any added noise signal amplitude to ensure the MOSFET stays properly biased. This OMS circuit allows you to accurately generate noise signals with bandwidths up to about 500 KHz. The bandwidth can vary depending on the DUT's input impedance and how well the OMS circuit was constructed. For best results, keep wiring and leads as short as possible to prevent oscillations.
As an example, the OMS circuit was implemented using an Agilent 33521A function/arbitrary waveform generator, N6762A power supply, E3630A dual-output power supply (to power the op amp), an IRFP150N N-type MOSFET, and an NE5534A op amp. The below scope capture shows the example OMS implementation generating a 200-mVpp 500-KHz sine wave onto a 5-VDC level. The same DUT load shown in the above diagram was used for this example, the 100-uF capacitor is electrolytic and the 1-uF capacitor is ceramic.
The AMS does not have the voltage or bandwidth limitations that the OMS has, but it does have a higher price tag. The AMS is shown in the below figure. The AMS consists of a power supply, FG/AWG or signal generator, wide-band or RF power amp and coupling capacitor section.
As an example, the AMS circuit was implemented using an Agilent 33521A function/arbitrary waveform generator, an N6762A power supply, an E&I 1040L RF power amplifier, and 10-uF and 0.1-uF capacitors in parallel (for the coupling capacitor section). A 1040L RF amp provided up to 400 W of output power, which was more than necessary, but I used what was available to me in the lab. Using the same load used in the OMS example, the AMS was used to create a single tone noise signal. The below scope capture shows a 1-Vpp 5-MHz sinewave was modulated onto a 10-VDC level.
One safety precaution you can take is adding blocking impedance between the power supply output and the coupling capacitor section. The blocking impedance can either be an inductor or a resistor. Its purpose is to protect the power supply from being damaged by the noise signal by adding higher series impedance before the power supply output. The inductor is ideal because it acts like a short to the DCV from the power supply, but presents high impedance to the AC noise signal. However, the inductor can cause large oscillations on the DC level if the DUT has a very dynamic current.
Using a resistor avoids the oscillations problem. Be sure to use the power supply’s sense leads to regulate the output voltage at the load to compensate for the voltage drop across the resistor. Also be sure to use a low-value resistor no higher than 1 Ohm. This is to ensure a low voltage drop across the resistor so you do not damage the sense lead circuit on the power supply.
PSRR testing is required on a wide range of devices today from smart phone chips to electronic assemblies for aircraft. Testing PSRR includes generating high-frequency noise signals on power supply levels. This task is challenging due to the high capacitance present at most power supply inputs. In this article we looked at two low-cost solutions for modulating noise signals on power supply levels. If you have any questions on the post send me an email and if you have anything to add use the "Comments" section below.